Respiratory Support/Long Term Ventilation after hours issues
This document is only valid for the day on which it is accessed. Please read our disclaimer.
This information is for staff dealing with children receiving long term respiratory support / ventilation outside of the Respiratory Department / Laboratory's ordinary working hours.
- Respiratory support / ventilation via the Respiratory Department / Laboratory is intended for long term use in children with stable requirements, not acute care. The Laboratory does not currently provide after-hours technical support or equipment.
- Children receiving community respiratory support / ventilation via the Respiratory Department should have a Respiratory Emergency Management plan on Concerto detailing their underlying condition, respiratory support, degree of 'dependency' and plans for both acute deterioration and equipment failure.
- Details of a child's respiratory support / ventilation equipment as supplied by the Respiratory Laboratory are detailed in the Physiologist E-note on 3M Chartview.This appears at the very bottom of their Chartview documentation by default.
- The on-call Respiratory Specialist is available 24/7 via switchboard, ext 21900 or 09 365 4330. Outside of PICU, they should be involved in any changes to respiratory support equipment (masks, circuits, device, addition of oxygen, change in settings).
Children receiving ventilatory support are diverse in aetiology, needs, dependency and support offered, making it difficult to generalise. The Respiratory Department typically classifies children into broad categories to aid planning and this should appear in the Emergency Management Plan on Concerto.
(continuous or sleep)
|These children are highly dependent on their ventilation. They will typically have backup equipment with them and if acutely unwell would be admitted to PICU by default.|
|Health support||These children (the majority) can typically go several days without ventilation without experiencing severe irreversible harm. They will not usually have backup equipment and technical issues can usually wait until the Respiratory Laboratory / Department's ordinary working hours. If acutely unwell it may be appropriate to discontinue, maintain or escalate their respiratory support. If support requires escalation consider whether they need PICU admission (default).|
|Fragile health support||These children aren't viewed as highly
dependent (can usually go without ventilation for a period) but for
a variety of reasons are particularly fragile.
They may have backup equipment and/or additional alarms. They may have very specific plans and specialist discussion is recommended.
|Palliative support||These children may be highly dependent but may have different therapeutic goals and very specific management plans. They may have backup equipment and additional alarms. They may have Advance Care Plans.|
- Appropriate action will depend on the child and the nature of their acute illness.
- Check with the child's Respiratory Emergency Management Plan.
- See categories above
- It may be appropriate to discontinue, maintain or escalate their ventilation. Caution is recommended where ventilation is maintained or escalated as this may be a change in indication and the child may now be best managed / monitored in higher intensity environments (i.e. IOA / HDU / PICU).
Admitting a child on respiratory support / ventilation
Starting ventilation acutely
High flow nasal prong therapy ('High flow' / Airvo) may be started acutely in ED or on ward as per prevailing Starship protocols. Other forms of ventilation (CPAP, BPAP, etc) are generally only started acutely within PICU.
Changing pressure settings
Only do this in consultation with the on-call Respiratory or Intensive Care Specialist. Each device is different. Equipment instructions can be found on the Starship L drive. This folder is open access (on ADHB network) - to access copy and paste L:\Groups\STARSHIP\Respiratory\Resp lab\Equipment instructions into your ADHB web browser. There may be specific recommendations regarding this in a child's emergency management plan.
Changing to different mask
Only do this in consultation with the on-call Respiratory or Intensive Care Specialist. Fitting a new mask requires specific skills. Changing a mask or an incorrect fitting may affect ventilation outcomes. Almost all masks used for community / ward Long Term Ventilation (LTV) are vented masks (holes for intentional leak). Using a non-vented mask (from PICU) will lead to rebreathing and clinical deterioration.
Equipment instructions can be found on the Starship L drive. This folder is open access (on ADHB network) - to access copy and paste L:\Groups\STARSHIP\Respiratory\Resp lab\Equipment instructions into your ADHB web browser. These documents are intended as aids for those already familiar respiratory support / ventilation. The setting up of ventilatory support by untrained staff is not recommended for dependent / fragile children and may lead to serious adverse events.
Replacing failed equipment
Children so dependent that they cannot go at least one night without support should have been supplied with backup equipment (interfaces, circuits, ventilator). Where this is not available, circumstances have changed and/or safety concerns exist, liaison with the on call Respiratory Specialist is recommended. In general admission to PICU/HDU for support on ICU equipment will be the most appropriate outcome until replacement equipment can be sourced and appropriately set up. Setting up equipment by untrained staff is not recommended and may lead to serious adverse events.
Supplemental oxygen can be added to all forms of ventilation. Oxygen must not be left running when a device is off (fire / explosion risk).
|CPAP Icon||Oxygen T-piece in circuit (at either the device or mask end)||0-15L/min|
|CPAP Sleepsense||Oxygen T-piece in circuit (at either the device or mask end)||0-15L/min|
|CPAP other||Likely via oxygen T-piece in circuit (at either the device or mask end)||0-15L/min|
|Oxygen T-piece in circuit (at either the device or mask end)||0-15L/min|
|Specific circuit (CLIMATELINE MAX OXY 36996) or via oxygen adaptor T-piece in circuit||0-15L/min|
|Resmed Lumnis||Specific circuit (CLIMATELINEAIR OXY 37357) or via oxygen adaptor T-piece in circuit||0-15L/min|
|Resmed Stellar||Oxygen port at rear (white adaptor
|Resmed Astral||Oxygen port at rear (white adaptor
0-6L/min in travel bag
|F&P Airvo||Oxygen port at side or rear||0-15L/min|
** If a child doesn't already have the correct circuit or adaptor, these may be found on Ward 26b or in the Starship Respiratory Laboratory (Level 3) - consult with the on call Respiratory Specialist
Did you find this information helpful?
- Date first published: 14 November 2018
- Date last published: 14 November 2018
- Document type: Clinical Guideline
- Services responsible: Paediatric Respiratory
- Author(s): Jacob Twiss, Tony Bell
- Editor: Greg Williams
- Review frequency: 2 years
SIGN UP TO RECEIVE GUIDELINE UPDATES
Subscribe below if you want us to let you know about new or updated guidelines
More From Starship
CareConnect TestSafe is a way for clinicians to get remote access to Starship clinical documents. Find out more...
Read about the governance process around the Starship Clinical Guidelines and how to format guidelines in development.